Thank you very much for the Excel document! It helped me a lot.
But I found an error:
Constant aa5 value -0.61893395 (H30) is wrong. It must be -0.61899395.

Hi Shane,
is there a formula could i used to compute ( a4 b4 a7 b7 c7 results of Nominal current Calibration ) I cant find any formula for this equation . i find it in may sensor calibration certificate. hope you can give me idea how can i compute.

Could you check out about the titles:
i. Above Zero Calculator 0.01 °C to 660.323 °C
ii. Below Zero Calculator -189.3442 °C to 0.01 °C
iii. K and not °K

Nguyễn Việt PhươngTemperature range from 660.323 to 961.78 need d coefficient but I didn't see it in your excel file?

Elmar BaumhöggerThank you very much for the Excel document! It helped me a lot.

But I found an error:

Constant aa5 value -0.61893395 (H30) is wrong. It must be -0.61899395.

Shane AlldredgeThanks Elmar. Correction made.

Tom HuganaHi Shane,

could you please add a comment to the R=>T conversion including cell B22. Is there any reference to this?

Alex ArellanoHi Shane,

is there a formula could i used to compute ( a4 b4 a7 b7 c7 results of Nominal current Calibration ) I cant find any formula for this equation . i find it in may sensor calibration certificate. hope you can give me idea how can i compute.

thanks,

AlexA

Ciro Alberto SanchezGreat work. ¡¡¡Congrats!!!

Could you check out about the titles:

i. Above Zero Calculator 0.01 °C to 660.323 °C

ii. Below Zero Calculator -189.3442 °C to 0.01 °C

iii. K and not °K

Ciro Alberto SanchezTo Alex Arellano:

You can found useful:

http://download.flukecal.com/pub/literature/3498460B_EN_w.pdf

https://us.flukecal.com/ITS-90-Reference-Function-Calculator