Title: 1551A or 1552A SCPI commands for Auto Time-Out.
Applies To: 1551A or 1552A Stik Thermometer
Problem Description: SCPI commands for Auto time-out are not listed in the manual.
The commands are as followed:
Returns the state of the automatic power off feature.
The response is OFF, or the setting 1 to 30 minutes.
CALibrate:DEVice:AUTO <val> [CAL:DEV:AUTO <val>]
Sets the state of the automatic power off feature.
The <val> parameter is ON, OFF, or the value 1 to 30 expressed in minutes.
If ON is selected the time is set to 30 minutes.