Note that the new version of the 700PCK software that is compatible with 64-bit operating systems is available for download at this link 700PCK 64-bit OS. This 700PCK software is required for use with these new COMPASS macros.
Download the sample COMPASS for Pressure Database named "700P CompSetP_xfer.mdb" from the link at the end of this article. It includes this equipment:
- Fluke 700P pressure modules; 15 psid and 5 psia gauge ranges
- PPC4 Pressure Controller
Note that the interface method in these setups does not require the flukeinterface.exe file like the previous COMPASS macros did.
Note that COMPASS must be closed and reopened after each test (or attempted test) when interfacing with 700PCK. If this is not done communications between COMPASS and 700PCK will not be established in the next test.
The updated macro names in this database all have 700P in them (instead of just 700) to designate that they are for 64-bit operating systems and the newer 700PCK software. For example, the new interface macro is now named "Fluke700P_GetPressure" while the old one was "Fluke700_GetReading". This is also to differentiate them from the older macros that required flukeinterface.exe. The old macros can be deleted from your database once you have confirmed that the new macros and tests work correctly. flukeinterface.exe is no longer required.
Once downloaded, you have two options to view, use or transfer the setups.
1. Open the COMPASS Database Maintenance Tool via the [Database], <COMPASS Database Maintenance Tool> menu path. For instructions to use this to transfer setups (Support Device, DUT, Test, etc.) to your database open the COMPASS Help File and type "Database Maintenance Tool" in the Search box.
2. Select this database via the [Database], <Setup> menu path. Click on the current database shown in the "COMPASS Configuration Database" box and then select the "700P_CompSetp April 2015.mdb" database. Note that you might have to add your COMPASS license to this database via the [Database], <License Management> menu path if your license is not automatically added to the database
You will have to edit existing DUT and Test Setups to use the new macros. DUT Setups need the new Interface macro "Fluke700P_GetPressure". Here's an example.
Edit the Test Setup to have the Zero test macro if you want to zero at the first point (or any point) in the test.
If you want to run an as-found verification run, adjust the module, and then run a as-left verification run you must specify that you want to run two Pressure Cycles (adjustment does not count as a cycle). Change "Pressure Cycle" to "2".
And in the Data tab choose the "Fluke700PCalibrationAdjust" macro as the active Test Event Macro. The Post Test Macro isn't necessary but may be in the future (see remaining issues below).
The default RS232 COM port is 1 (one). There are two ways to change this.
1. When initializing the test the COM port that the 700PCK hardware is on can be selected by entering the COM port number in the "Parameter ID" field. In the image below it is on COM2. If the Parameter ID field is left blank then the default port will be used from the macro. If you are have more than one DUT you must enter the COM port for the second, third, fourth, etc. DUTs in the Parameter ID field.
2. The COM port can also be changed in the Interface macro "Fluke700P_Get Pressure" by changing the number in the line that reads "defaultPort = 1" to the actual COM port.
Update, 31 August 2015:
The database has been updated with new macros which include a pair of 700P/750P macros for use with a deadweight tester or a non-automated pressure controller/calibrator.
Update, 22 September 2015:
Edited the Interface macro "Fluke700P_GetPressure" so that the default port specified in the macro now works if Parameter ID is left blank.
1. COMPASS must be shutdown (closed) after each run because it's not releasing the COM port(s)
If you have questions contact Fluke Calibration Pressure Technical Support at the below link or by calling +1.877.355.3225