In COMPASS for Pressure the ID of the Q-RPT of the reference device is not saved in the data file. For example if you have a PPC4 A7Ms SN 786 pressure controller with an attached RPM4 A2Ms/G200K pressure monitor and you AutoRange to 1.4 MPa (200 psi) the A2Ms will be used. But the only identifier of the reference device in the data file will be that it is PPC4 SN 786, Final Output Max 1.4 MPa.
This Test Macro will save the Q-RPT identifier (X1L) in this example to StringData1 in the datafile for each data point. Other parameters are available to be saved also (Autorange range, Autorange span, Q-RPT range, QRPT span, QRPT mode)
You can then choose to put this information on your calibration report, leave it in the data file for traceability or use for other calculations (measurement uncertainty, etc.).
Link to the file is below the image.
Contact Fluke Calibration Pressure Technical Support at Pressure Support or by phone at +1 877 355 3225