A Pressure Measurement Module Calibration Kit (Calibration Sled) is available if you would like to calibrate PM600 or PM200 modules outside of the 6270A or 2271A pressure controller. The kit includes a mains power supply adapter, a null modem cable (DB9-F to DB9-F), a null modem USB to RS232 adapter cable (and driver CD), and a plug for the REFERENCE port (for use when calibrating PM200-A100K, PM200-A200K or BRM600-BA100K modules). The TEST and REFERENCE pressure ports are 1/4 inch NPT. COM settings are: baud rate 9600, parity none, data bits 8, stop bits 1, handshake: none. Commands sent to the kit should be terminated with CR and LF. The replies from the kit are terminated with CR. The reply to a *IDN? query is "Fluke, Model #, Serial #, etc”. See other commands and descriptions below.
Part Number: 4456687
Description: PMM-CAL-KIT-20M, PRESSURE MEASUREMENT MODULE CALIBRATION KIT, 20 MPA (3000 PSI)
Contact the Fluke Calibration Orders Team by email at FCAL Orders Team or by calling +1.877.355.3225
Note: Commands below are in quotes for clarity of what the full command is. Don't send quotation marks.
To read the current pressure send "VAL?" command. The reply will be the value and units in kPa.
To read the serial number send "*IDN?" command. Reply is "Fluke, Model #, Serial #, software version, model, memory. Example reply is FLUKE,PM600-A20M,9990101,1.00,PM600,64K
Whenever you make changes to settings you must send “CAL_STORE” to save to memory. If not, and power is cycled, the value will not be stored.
To read the pressure unit send "PRES_UNIT?" command. The return is the pressure unit.
To set the pressure unit send "PRES_UNIT x" where "x" is KPAL (for kPa), PSI or BAR
C0 (Pressure Adder):
To read C0 send “OFFSET_SET?” The return is in psi.
To set C0 send “OFFSET_SET x” where “x” is a decimal or integer in psi.
C1 (Pressure Multiplier):
To read C1 send “GAIN_SET?” The return is a unit less decimal.
To set C1 send “GAIN_SET x” where “x” is a decimal (multiplier).
To read ZOFFSET send “UCOEF_SET?” The return is in psi.
To set ZOFFSET send “UCOEF_SET x” where “x” is a decimal or integer in psi.
To read cal date send “CAL_DATE?” The return is a string.
To set cal date send “CAL_DATE mm/dd/yy”
To save changes to memory send "CAL_STORE" command
Whenever you make changes you must send “CAL_STORE” to save to memory. If not, and power is cycled, the value will not be stored.
The PMM Calibration Kit is just like a bay in the 6270A or 2271A pressure controller. Remove the Pressure Measurement Module from the bay in the 6270A or 2271A pressure controller and insert it in the same way into the PMM Calibration Kit.