Dumb 725 Question
As I am clearly awesome, I will chalk this silly issue up to the meter being dumb and not me. Because, well, come on, it's me! I have a Fluke 725, Firmware 2.xx that I am trying to send a source value to. Meter is set to mV Source and am trying to set it to 45mV. So my commands are as follows. 2.001 PORT [CLR]m 2.002 PORT 45 That always sets the meter to 4mV output. I have tried about every ASCII construction I can think of: 2.002 PORT 45.00 2.002 PORT 4,5 2.002 PORT 4,5,0,0 2.002 PORT 126.96.36.199 ...and others. Still, only the 4 takes. I thought I was onto something playing with the inter-byte delay until I read in the Met/Cal manual that this only works with IEEE constructions. So... What am I missing?
Please sign in to leave a comment.