Using the Fluke 5730A in 5700A or 5720 Emulation Mode
I am using our new 5730A in emulation mode in place of our 5700 and 5720 calibrators in MetCal 7. I have managed to get it to work but am having occasional timeout errors that causes the 5730 to lock up and require a hard powerdown. Does anyone have any tips on how to correctly set up the configuration file and the NI Max communications which might be the cause of these errors?
Please sign in to leave a comment.