HP 34401A automatic calibration using Fluke 9100E
Please help solve the problem:
there is such a simple code
1.001 PORT [P9600,N,8,2,D][OTERM LF][TERM LF][T10000]
1.001 PORT SYST:REM;*CLS;*RST
1.002 PORT SYST:REM;*CLS;*RST;:TRIG:SOUR BUS
1.003 PORT FUNC "VOLT:DC";:VOLT:DC:RANG 100mV;RES 0.0001mV
1.004 9100 20 100mV 0.0085U 2W
2.001 TARGET -m
2.002 CALL Sub HP 34401A Get Reading RS-232
2.003 MATH MEM = MEM / 1E-3
2.004 MATH MEM = CRND(MEM, 0.0001)
2.005 MEMCX 100 100.0mV 0.0085U
On line 1.004, a window to enter the enter values with the keyboard.
How to disable it, that would be read out testimony from 34401A?
Please sign in to leave a comment.