Configuration of Fluke 8508A with Fluke 5500A
Can anyone clarify the following:
1. How to do the configuration for Fluke 8508A -Reference Multimeter with Fluke 5500A through IEEE 488 GPIB cable.
2. What is the procedure to be followed for Fluke 8508A “Function Zero” & “Range Zero”?
3. How to set the Fluke 8508A - Reference Multimeter to default Mode. As the display indicates that the massage as “Calibration Corrupt-Default “on the display in normal operation and I am unable to retain to the default mode.
Please refer the below photo as illustrate the massage “Calibration Corrupt-Default “
Your earlier response will be highly appreciated,
Please sign in to leave a comment.