8846 time out
Since updateing to metcal 8.3 all programs that have an 8846A as the standard now time out, program puts the meter in the right mode/range but does not read back the value and time out. Procedures worked before and tested same procedure susbstituting an 8508A for the 8846A and the procedure worked . Is this a problem with the 8846A firmware or with MetCal 8.3? I have tried several 8846A meters on the bench.
Please sign in to leave a comment.